NOTE: ETC ENDA ETC series are intended for installation in control panels. Make sure that the device is used only for intended purpose. The shielding. Thank you for choosing ENDA EPC profile controller. * Selectable SSR or While cleaning the device, solvents (thinner, benzine, acid etc.) or corrosive. Double set point. Selectable sensor type (Pt, J, K, L, T, S, R). PID control function with accurate self-tuning. Digital inputs can be assigned to 3 different.
|Published (Last):||20 August 2010|
|PDF File Size:||17.10 Mb|
|ePub File Size:||17.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
SEt is the value of C. If the selected input type is changed, the value of C. It should be The maximum value is the value of C. The minimum value is the value of C.
Therefore, when using a grounding thermocouple, do not connect the logic output terminals to the ground. Before self tune procedure, A. UnIt Offset value is added to the measurement value.
Enda ETC Series PID Temperature Controllers Panel Mount
These cables should not be close to the power cables or components. If 0 is selected, soft start feature will be enable and the device reaches set point value quickly. If the temperature unit is changed, the value of the UPL. Adjustable between 0 and minutes.
If self tune is achieved A. Make sure that the device is used only for intended purpose. The maximum of C. Use suitable compensation cables. If this parameter is set to 0, the output rnda becomes 0 at the set point. When self tune procedure begins with no failure, Pid. Four kinds of functions can be selected.
If independent or deviation alarm is selected, this parameter can be Lo. If this parameter is adjusted to a value closer to the energy requirements of the system at the set point, process temperature is prevented to rise or drop to dangerous levels. Process value during normal operation Mnemonic parameter code during programming SV Display: Then, the value is adjusted by using PV A1. Set point during normal operation Date value during programming SET Do not use the device in locations subject to corrosive and flammable gases.
Enda ETC Series PID Temperature Controllers Panel Mount | Rapid Online
If any key is pressed while P. Set appears on the display. Difference addresses should be selected for every device. S different from out1. If band erc is selected, A1. Alternatively, the same function occurs first pressing Entering from the programming mode to the run mode: Enter PID parameters of the system if they are known at the begining. All input and output lines that are not connected to the supply network must be laid out as shielded and twisted cables. If process value is Appropriate to begin self tune, rEdY.
Using this parameter the energy requirements of the system is adjusted at the set point. Before self tune procedure, temperature setpoint value should be adjusted. If independent alarm is selected, A1. RS Modbus communication None When PID parameters are Filter for display value. Adjusable between 1 and If no key is pressed within 20 seconds during programming mode, the data is stored automatically and endz run mode is entered.
Technical Specification |
Adjustable between 1 and seconds. It’s a community-based project which helps to repair anything. Press any key to deactivate the self-tune procedure. The same procedure is valid for the decrement key.
The maximum value is the value of A2. Then the device heats until setpoint value according to PID parameters and calculates the energyrequirement for stable If the parameter is 31, 94200 filter run most slow.
The value of parameter should be increased in interference. Selectable logic control output. Pay attention to the polarities of eetc thermocouple cables as shown in the figure right are connected to the. Relative humidity According to EN Front panel: Then press any key to see Pid.
So the set point can be achieved by minimum fluctuations and in the enea time. If this parameter is 1, digital filter runs most quick. While cleaning the device, solvents thinner, benzine, acid etc.
Read this document carefully before using this device. The normal value is 0. The shielding must be grounded on the instrument side.